Turning on HRRL
Turn on water pump for cooling system
There are two pumps, use them alternatively when run multiple days.
Turn on ISA water skid system
Turn on breaker and then press start.
Turn on Modulator
Unlock padlock on SCR Enclosure on Moculator and turn switch to on. (This turns on 480 V AC for HVPS)
Turn on filament
Press Startup on PanelMate touchscreen locate at control rack
Touchscreen at control rack:
Wait for 900 s count down.
Turn on magnet power supplies
Check if the modulator is on
if the two lights at the corner of the box is on, then modulator is on.