

Line 28: 
Line 28: 
 <math>V_2</math>  with Ribbon Cable   <math>V_2</math>  with Ribbon Cable 
   
−  ==Pictures for pulse's shape change==  +  ==Pictures about change of pulse shape== 
   
 { border="1" cellpadding="20" cellspacing="0"   { border="1" cellpadding="20" cellspacing="0" 
Revision as of 22:15, 19 January 2011
Go Back
Experimental Set Up
setup without Ribbon Cable 
setup with Ribbon Cable



Measurements
Pulse width 
[math]V_1[/math] 
[math]V_2[/math] 
Attenuation, % 
Comments

[math]100\ \mu s[/math] 
[math]5.0\ V[/math] 
[math]4.5\ V[/math] 
10 % 

[math]10\ \mu s[/math] 
[math]5.0\ V[/math] 
[math]4.2\ V[/math] 
16 % 

[math]1\ \mu s[/math] 
[math]5.0\ V[/math] 
[math]3.0\ V[/math] 
40 % 
see pictures about change of pulse shape

[math]100\ ns[/math] 
[math]5.0\ V[/math] 
[math]2.8\ V[/math] 
44 % 
see pictures about change of pulse shape

[math]30\ ns[/math] 
[math]4.0\ V[/math] 
[math]1.8\ V[/math] 
55 % 
see pictures about change of pulse shape

[math]V_1[/math]  without Ribbon Cable
[math]V_2[/math]  with Ribbon Cable
Pictures about change of pulse shape
[math]1\ \mu s[/math] pulse without Ribbon Cable 
[math]1\ \mu s[/math] pulse with Ribbon Cable



[math]100\ ns[/math] pulse without Ribbon Cable 
[math]100\ ns[/math] pulse with Ribbon Cable



[math]30\ ns[/math] pulse without Ribbon Cable 
[math]30\ ns[/math] pulse with Ribbon Cable



Go Back