Difference between revisions of "CLAS12R1 CosmicTestStand"
|Line 37:||Line 37:|
Revision as of 18:08, 19 December 2012
Two scintillators are placed on top and below the drift chamber
These scintillators have PMTs on both ends allow an initial position to be determined according to the time difference between the PMT hits.
Below is the calibration for Detector M and Detector G
HV(M1) = -1602 Volts HV(M2) = -1750 Volts
HV(G1)= -1700 Volts HV(G2)= -1650 Volts
M1 = TDC80 M2 = TDC81 G1 = TDC84 G2 = TDC86 Trig = TDC 95
M1 = TDC112 M2 = TDC113 G1 = TDC116 G2 = TDC118 Trig = TDC 127
Trigger Matching is ON resolution = 100 ps